- Benchtop High Speed Centrifuge
- Floor high speed centrifuge
- High speed refrigerated centrifuge
- Tabletop Low Speed Centrifuge
- Low speed refrigerated centrifuge
- Free-standing Low Speed Centrifuge
- Laboratory centrifuges
- Medical centrifuge
- Automatic decapping centrifuge
- Oil-test centrifuge
- Large capacity centrifuge
- Hematology Centrifuge
- PRP centrifuge
- Refrigerated Centrifuge
- Blood Bank Centrifuge
- Mini centrifuge
- Special centrifuge
- Centrifuge rotors
- Electronic Balance
- Vacuum Centrifugal Concentrator
- Moisture Analyzer
The centrifuge is an often use equipment in the laboratory, by the high-speed rotating of rotor produced the centrifugal force, making the matter with different density, dimension separated, it’s to achieve the goal of preliminary Separation and purification which is an essential instrument for the lab. And it include Free-standing High-speed Refrigerated centrifuge, Table-type Low-speed centrifuge, etc.
If misoperation of the centrifuge, it will make the accident even lead to the tragedy.
How to correctly use the centrifuge, prolong its useful life and reduce the accident happen?
A. Require of centrifuge's placement.
a.We as far as possible to choose the places which is dry, and not direct sunlight for the centrifuges.
b.No sundries around the centrifuge, and distance of the walls, baffle and so on (airtight and poor heat) at least more than 10 cm.
c.Let the centrifuge alone in a room as far as possible, and never put around of organic reagents, in-flammables of it.
B. The dangers of the centrifuge.
When Centrifuge beginning work, is a very high speed rotor, and is dangerous very much. If the high speed and ultra-high speed rotating rotor flying out, almost breakdown anything.
C. the matters need attention of using centrifuge
According to the experiment use of the centrifugal tube, the liquid containing organic reagent it must use special centrifugal tube;
b.Choose the right turn, and never using the revolving speed more than the highest speed of rotor was marked;
c. It is very important to correct use of the centrifugal tube and the rotor, and it’s especially correct use the symmetry placement of centrifugal tube and balancing of weight;
d. The centrifuge tube and rotor can withstand solution's maximum weight is centrifuge tube 3/4 of the volume.
- PREVIOUS：the difference between Centrifuge and scillating screen
- NEXT： About Laboratory centrifuge